Dendritics Gemscale-50 Owners Manual Page 3
Place scale on
stable, flat, horizontal surface.
Press Lever to open.
Press Button to turn
on, Reads <188.88>
Calibration Weight on Pan.
weight and start weighing.
Press Button once
briefly to zero Display.
Press Button down
until Display shuts off or scale will automatically shut itself off in
replacement when scale will not give a reading.
Compartment door on the bottom of the scale.
Take a thin flat
object such as the flat end of stone tweezers. Put it into the battery door
groove and press inward and up. The battery door will flip open for you.
Replace with a 9 Volt
Alkaline for 300 operating hours or a 9 Volt Heavy Duty for 200 operating
Put the battery door
hinges in place. Press battery door down until it "clicks" and shuts securely.
<Err> Weight too
<-Err> Weight below
zero, weighing pan missing.
<----> Waiting for a
stable reading. Use draftshield.
indications for two minutes. Needs factory
Images and webpage designs © 2001-2018 jb and Dendritics Inc. [-]